Medical electrical equipment. Part 2-41: Particular requirements for basic safety and essential performance of surgical luminaires and luminaires for diagnosis (IEC 60601-2-41:2009)
Clause 1 of the general standard1) applies, except as follows: 201.1.1 *Scope Replacement: This particular standard applies to the BASIC SAFETY and ESSENTIAL PERFORMANCE of SURGICAL LUMINAIRES AND LUMINAIRES FOR DIAGNOSIS, hereafter referred to as ME EQUIPMENT. This particular standard does not apply to – headlights; – endoscopes, laparoscopes and their light sources, which are covered by IEC 60601-2-18; – luminaires used in dentistry, which are covered by ISO 9680; – luminaires for general purposes, which are covered by IEC 60598-2-1 and IEC 60598-2-4; – luminaires dedicated to therapeutic purposes; – special purpose lights with different conditions of use such as UV lights for dermatological diagnosis, slit lamps for ophthalmology, lights for surgical microscopes and lights for surgical navigation systems; – lights connected to surgical instruments; – luminaires of an emergency lighting, which are covered by IEC 60598-2-22. NOTE See also 4.2 of the general standard. 201.1.2 Object Replacement: The object of this particular standard is to establish particular BASIC SAFETY and ESSENTIAL PERFORMANCE requirements for SURGICAL LUMINAIRES and LUMINAIRES FOR DIAGNOSIS as defined in 201.3. 201.1.4 Particular standards Replacement: In the IEC 60601 series, particular standards may modify, replace or delete requirements contained in the general standard and collateral standards as appropriate for the particular ME EQUIPMENT under consideration, and may add other BASIC SAFETY and ESSENTIAL PERFORMANCE requirements. A requirement of a particular standard takes priority over the general standard. For brevity, IEC 60601-1 is referred to in this particular standard as the general standard. Collateral standards are referred to by their document number. The numbering of clauses and subclauses of this particular standard corresponds to that of the general standard with the prefix “201” (e.g. 201.1 in this standard addresses the content of Clause 1 of the general standard) or applicable collateral standard with the prefix “20x” where x is the final digit(s) of the collateral standard document number (e.g. 202.4 in this particular standard addresses the content of Clause 4 of the 60601-1-2 collateral standard, 203.4 in this particular standard addresses the content of Clause 4 of the 60601-1-3 collateral standard, etc.). The changes to the text of the general standard are specified by the use of the following words: "Replacement" means that the clause or subclause of the general standard or applicable collateral standard is replaced completely by the text of this particular standard. "Addition" means that the text of this particular standard is additional to the requirements of the general standard or applicable collateral standard. "Amendment" means that the clause or subclause of the general standard or applicable collateral standard is amended as indicated by the text of this particular standard. Subclauses, figures or tables which are additional to those of the general standard are numbered starting from 201.101. However, due to the fact that definitions in the general standard are numbered 3.1 through 3.139, additional definitions in this standard are numbered beginning from 201.3.201. Additional annexes are lettered AA, BB, etc., and additional items aa), bb), etc. Subclauses or figures which are additional to those of a collateral standard are numbered starting from 20x, where “x” is the number of the collateral standard, e.g. 202 for IEC 60601- 1-2, 203 for IEC 60601-1-3, etc. The term "this standard" is used to make reference to the general standard, any applicable collateral standards and this particular standard taken together. Where there is no corresponding clause or subclause in this particular standard, the clause or subclause of the general standard or applicable collateral standard, although possibly not relevant, applies without modification; where it is intended that any part of the general standard or applicable collateral standard, although possibly relevant, is not to be applied, a statement to that effect is given in this particular standard.
: 11 390,0 Ft
(papír , PDF-fájl-letöltés)
Jelenleg nincsenek utódok